The Eko CORE is an electronic stethoscope that enables amplification, filtering, and transmission of auscultation sound data (heart, lungs, bowel, arteries, and veins). The CORE is a small attachment that is located between the tubing and chest piece of a stethoscope.
The Eko CORE is available as part of a fully-assembled 3M™ Littmann® CORE Digital Stethoscope.
What’s included in your box
Fully assembled 3M™ Littmann® CORE Digital Stethoscope
Additional set of 3M Littmann Snap Tight Soft Sealing Ear Tips (Small)
Nonchill bell sleeve
USB charging cable
Getting Started Guide
Watch the video
Turn on and turn off the CORE
Here’s how it works:
Depress the power slider to move the switch from the OFF to the ON position.
“ON” is when the toggle is flushed with the surface of the volume buttons. You will see a white blinking light to indicate that the CORE has been turned ON. You can now hear amplified sounds through the stethoscope. The CORE is also ready to pair with a mobile device and transmit digitized sounds.
OFF is when the toggle is protruding from the surface of the volume buttons. The white light will disappear and you can now hear unamplified sounds through the stethoscope earpieces.
The CORE has a 5-minute standby mode to conserve battery life. The white light disappears when the CORE goes into standby mode and the toggle remains flat with the surface of the volume buttons (the “ON” position). To power the CORE back ON, push the toggle to the OFF and then the ON position. The white light will appear to indicate the device is ON.
Adjust the volume
The CORE’s sound level can be amplified in 7 increments up to 40X amplification over an unamplified, acoustic stethoscope when listening with the stethoscope earpieces. Change the volume level by clicking the plus (+) and minus (-) volume buttons on the side of CORE.
If you're using headphones to listen to the sounds through your mobile device, you’ll need to adjust your phone’s volume to change the CORE’s volume.
Changing the volume on the CORE does not affect the volume of a recording on the Eko App running on your mobile device. The recording volume must be adjusted in the App or on the phone itself.
Charge the CORE
When the yellow light starts blinking on the CORE, the battery in the CORE needs to be charged.
Insert the included micro USB cable into the USB port on the CORE and plug the other (USB-A) end into a UL-certified USB wall charger (max 5 volts). The LED turns solid yellow, signifying that it's charging. The LED changes to solid green when the device is fully charged.
To fully charge the battery, it should take about 2.5 hours. When fully charged, the CORE should last for about 8 hours of continuous use. We recommend turning the CORE OFF between each use to maximize battery life.
CORE will not turn on while it’s plugged in and charging.
Care and cleaning
The stethoscope and CORE should be cleaned between each patient's use. All cleaning instructions pertaining to the original stethoscope apply.
Under normal conditions it's unnecessary to remove the CORE from the stethoscope tubing for cleaning. All external parts of the hardware can be cleaned with 70% isopropyl alcohol wipes.
DO NOT immerse the device in any liquid or subject it to any high-pressure/autoclave sterilization processes.
If it becomes necessary to remove the CORE digital attachment from the rest of the stethoscope, pull the stethoscope tubing off of the metal stem of the attachment on both ends of the attachment. You may need to pull firmly to detach the components. Wipe all parts of the stethoscope clean with 70% isopropyl alcohol wipes including the CORE’s surface, stethoscope tubing, tubing connector, and chest piece. Reassemble the stethoscope by reinserting the metal stems into the stethoscope tubing as before.
Best practices for audio quality
Please follow the steps below to achieve the best quality auscultation when using your CORE and stethoscope earpieces:
If possible, perform auscultation in a quiet environment. While the CORE has active noise cancellation, some environmental noise may still be audible during auscultation.
Use the bell to best listen to low frequency sounds like heart sounds. Use the diaphragm to best listen to high frequency sounds like lung sounds.
Hold the diaphragm firmly against the chest. Avoid any motion. Beware of pressing too tightly, as that can muffle sounds.
Download PDF manuals
Date of Issue September 5, 2022